How does Hubble compare?
Posted By Emily Lakdawalla
2009/09/16 06:13 CDT
Apologies in advance for the number of acronyms in this post. There's no way to get around them.
Last week when the new, post-servicing-mission-4 capabilities of Hubble were unveiled, I kept asking myself: How do Hubble's new capabilities compare to what it could do before the servicing mission, and how do they compare to what we can do from the ground? I fired off some questions to Heidi Hammel, and did some research on my own within the Hubble instrument description documents.
There's four main Hubble instruments I was interested in comparing with each other and to Earth:
- The Wide Field and Planetary Camera 2 (WFPC2), an ultraviolet- to very near-infrared imager, Hubble's workhorse from 1993 to 2009;
- the Near-Infrared Camera and Multi-Object Spectrometer (NICMOS), a near-infrared imager, installed in 1997, which has had periodic problems with its cooling system and was most recently offline from November 2008 to August 2009;
- the Advanced Camera for Surveys (ACS), an ultraviolet- to very near-infrared imager, in use since 2002 but with a hiccup in 2007, mostly repaired in May 2009; and
- the newly-installed Wide Field Camera 3 (WFC3), which replaced WFPC2 in May 2009 and was also intended to largely supplant NICMOS.
Here's the executive summary: what WFC3 brings to the party is a killer combination of large field of view and pretty high resolution. It's an improvement on WFPC2 in almost every way. It improves marginally on the resolution of the largest-field-of-view optical images that Hubble was capable of before (ACS Wide Field camera), and it really beats the pants off of the field of view of NICMOS, though at lower resolution than NICMOS can manage. (So it's a good thing that they're now reporting progress on getting NICMOS' cooling system working again.) And although Earth-based instruments at the Keck, VLT, and Gemini observatories can beat Hubble resolution using adaptive optics, that's only at near-infrared wavelengths; shorter than 800 nanometers (and especially in the ultraviolet) there's nothing on Earth that can get sharper images than Hubble.
I spent a couple of hours this morning noodling around the Space Telescope Science Institute website reading instrument descriptions. I came up with the following diagram that compares the capabilities of the different instruments. The top row (above the dashed line) compares fields of view -- how much of the sky each one takes in. (For comparison, the Moon at 30 arcminutes or 1800 arcseconds across would be about twice the width of the whole diagram.) The bottom row (below the dashed line) compares pixel sizes. Ideally you'd like to maximize field of view and minimize pixel size.I took these diagrams directly from the instrument descriptions. The skewed appearance of the ACS fields of view is real -- it has something to do with the optics within Hubble. Also real is how some of the detectors on some instruments have overlapping fields of view (like those in WFPC2 and WFC3), while others (like those in NICMOS) are spread out on the sky, intended to be used totally independently.
Here's a table summarizing the same geometric information, plus info on detector size and wavelength sensitivity.
|WFC3 UVIS||162 x 162||0.04||2@ 2051 x 4096|
(35-pixel gap in between)
|WFC3 IR||123 x 136||0.13||1014 x 1014||850-1700|
|WFPC2 PC||34 x 34||0.046||800 x 800||115-1100|
|WFPC2 WF2, WF3, WF4||150 x 150 (L shaped)||0.1||Three 800 x 800||115-1100|
|ACS WFC||202 x 202||0.05||Two 2048 x 4096||350-1100|
|ACS SBC||34.6 x 30.1||0.034 x 0.030||1024 x 1024||115-1700|
|ACS HRC||29 x 26||0.028 x 0.025||1024 x 1024||170-1100|
|NICMOS NIC1||11 x 11||0.043||256 x 256||800-1800|
|NICMOS NIC2||19.2 x 19.2||0.075||256 x 256||800-2450|
|NICMOS NIC3||51.2 x 51.2||0.2||256 x 256||800-2300|
ow, what I really wanted to be able to do was to post a similar diagram, extending the comparison to Earth-based observatories. But after spending part of the morning trying to understand all the different variables and how they relate to each other, I didn't think I could. Here's what I can tell you. I asked Heidi Hammel to tell me which Earth-based instruments were most appropriate for comparison to Hubble. She gave me the names of three instruments that work on major scopes equipped with Adaptive Optics (AO). AO corrects Earth-based telescope images in real-time, as light is falling on the detectors, for the distortions caused by Earth's moving, shimmering atmosphere. They are:
- NIRC2 on the Keck II telescope on Mauna Kea, Hawaii;
- NaCo on the Very Large Telescope (VLT) at Cerro Paranal, Chile; and
- the NIRI imager with ALTAIR adaptive optics system at Gemini North Observatory, also on Mauna Kea.
Why not just pick a few interesting objects out there in space and actually look at some comparable images? Nancy Atkinson over at Universe Today already did that for some Hubble targets, though she didn't specify which cameras the before and after images came from. I searched for some things to compare between the newly released Hubble images and the Earth-based ones that Heidi suggested I check, but I couldn't find anything comparable. I believe that the reason for that is the newest Hubble releases were selected in order to showcase the broad field of view of the new Wide Field Camera 3, and none of those Earth-based imagers holds a candle to WFC3 (or ACS's Wide Field channel, for that matter) in field of view.
In the end I feel like this was not a very productive use of my day! I guess what I've learned is that there isn't an easy answer to the question of how all these facilities compare. Which one you'd like to use depends upon a complicated game, trading between resolution, fields of view, wavelength sensitivity, the kinds of filters available (something I didn't even get into here), and how hard it is to get a proposal approved for time on the instrument of your choice.
With that, I'd better quit, and move on to something else! Hopefully tomorrow I'll have more to show from a day's work.