Some interesting information in a Nature story about Phoenix
Posted by Emily Lakdawalla
11-12-2008 16:09 CST
I just came across a long article in Nature on Phoenix, by journalist and sometimes blogger Eric Hand, that contained two tidbits of information about the mission that I had not previously been aware of.Both have to do with the Thermal and Evolved Gas Analyzer, or TEGA, half of Phoenix' package of analytical instruments, which suffered so many frustrating problems during the mission. One item that Eric dug up had to do with the manufacturing error that caused the failure of most of the doors on TEGA to open properly. Eric reports that this problem had actually been noticed before launch:
One of the differences between the TEGA on Phoenix and that on [Mars Polar Lander] was a thin retracting cover to keep the instruments from being contaminated by any stowaway microbes from Earth. [TEGA instrument head Bill] Boynton and his team had noticed, on a test version of TEGA, that the brackets at the bottom of this cover were just a hair's width too big, and as a result obstructed the doors. They sent revised designs for the cover to the manufacturer, Honeybee Robotics of New York. New parts were delivered and installed. But Honeybee had made the new parts using the original flawed designs -- and nobody in Tucson checked them. "They should've caught it and we should've caught it, but neither of us did," says Boynton, ruefully.Ouch.
Ouch ouch ouch. That's a painful revelation.
The other interesting tidbit of information was that, about a month into the mission, NASA headquarters stepped in to issue orders to the operations team on how to use TEGA. NASA was nervous about all the problems that had cropped up, so, Eric wrote, "At the end of June, word came down that the Phoenix team was to treat its next TEGA sample as its last, and to go after a sample of rock-hard ice before it did anything else. The Tucson team had lost its autonomy." If you read this blog throughout the Phoenix mission you'll recall what followed; it's all detailed in my sol-by-sol summary of the mission. They expended more than a month of precious highest-power mission time to try to get an ice sample into the instrument, an effort that caused the team to put off doing other types of analyses, and which ultimately failed.
Strong reporting by Eric Hand; read his full article here.
Or read more blog entries about:
Fifteen years ago, Society members and passionate space advocates like you helped save the Pluto mission. Now we can do the same for missions to Europa and Mars.
Join over 26,400 people who have completed their petition and consider a donation to support advocacy efforts.